Objectives Liquid medications provide an alternative to splitting pills and dosages by measuring an amount of liquid rather than crushing tablets or opening capsules. Special attention should also be paid to specific patient groups with swallowing difficulties or requiring enteral feeding administration. Liquid medicines are also often used to ease withdrawal symptoms for people suffering from addiction. Nevertheless, filling liquid medication cups with the right medication and precise doses may be difficult for healthcare professionals. The Nooddis (‘Nominative Oral Dose Dispenser’—Pierre Lôo Hospital, France and Packinov, France) is a new robotic system for the automated filling of single dose liquid medications. Since the performance of such a complex piece of equipment depends on compliance of the service provider to our building guidelines, the process performance verification is a necessary prerequisite before starting routine production.
Methods The performance of the Nooddis robot (accuracy, precision, and tapering calculation) and its ability to fill medicine cups was evaluated with 18 different liquid medications using an automatic in-line checkweigher. Microbiological testing was also performed.
Results 648 sealed cups were prepared for qualification. The filling accuracy was within the limit of ±10% from 75 µL to 21.25 mL. The repeatability (% relative SD (%RSD) 0.05 to 4.93) and intermediate precision (%RSD 0.01 to 6.59) were validated for all preparations. All medicine cups met the requirements of USP and European Pharmacopoeia acceptance criteria for microbiological quality. Automated tapering calculations allowed for easy production of daily doses for the tapering periods chosen.
Conclusion Since the system met the required quality standards, the Nooddis robot, with automatic in-line tapering system, is regarded as an accurate technology that can fill the exact amount of liquid oral medication in single dose cups. This may promote closer monitoring, which supports medication tapering as well as medication misuse prevention. With a packaging cost similar to current unit dose cup systems, it is a relevant alternative to fractioned or crushed tablets, as well as opened capsules. Further developments for some sterile liquid medications are yet to be studied.
- validation studies
- liquid medication
- automation, packaging
Data availability statement
All data relevant to the study are included in the article or uploaded as supplementary information. Not applicable.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.